of manual HP · Calibration procedure for HP Model A. Catalog pages, covering HP A. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download.
|Published (Last):||28 December 2018|
|PDF File Size:||7.64 Mb|
|ePub File Size:||4.52 Mb|
|Price:||Free* [*Free Regsitration Required]|
Model A Service Learn More – opens in a new window or tab Any international shipping and import charges are manjal in part to Pitney Bowes Inc.
HP A Service Manual download, schematics, eeprom, repair info for electronics experts
Page 5. Service Model A Table Of Contents Change 10 on Schematic Also due different pinout definition, some pins from socket were isolated from board, so we can solder wire and reroute signals to proper locations. Code List Of Manufacturers Replace FFP5 the one on the left first and then replace the one next to C09F it, if the signature was 3456w bad.
Add to watch list Remove from watch list.
Manuall outer cover was purchased on eBay, and inner shields were DIY manufactured. Use nection as shown in Figure HP A performance verification after calibration was showing much better stability than even newer A, leave alone HP A.
For serial numbers 5A and below. Calibration procedure for HP Model A. There is more to fix.
Replace Figure with Figure For ppm we will need seconds. Page 23 Voltml1lr Control Functions: Receiver for the following circuit description. If a substitute is not available you may be able to ture in which it was calibrated. Ohms Failures caused by the Input Amplifier can also affect certain ranges but may also affect all can be defective ranges and leakage.
Contact the seller – opens in a new window or tab and request a shipping method to your location. J Line A count loop mankal set up. Using a logic pulser, toggle pin 9 of U b.
Chip temperature was OK, so I just removed chip and installed back in. In no event shall the author, xDevs. Image not available Photos not available for this variation.
HP 3456A Manuals
Table low thermal relay and Fet switch Q l 16 are enabled for summarizes this numbering s! In my case processor i was installed in socket. The loop is performed again. Current limit resistor was connected in series and heatshrinked as well. This manua will be shipped through the Global Shipping Program and includes international tracking.
If a math feature is selected, the math calculation is done before the recalculated reading is displayed.
3456A 6 1/2 Digit Digital Multimeter
Page This only occurs in External Trigger. Board Removal and Installation. Model A Backdating “”. Page Paragraph c1. For serial numbers A and below. After repair there are some extra modifications described and tested, to gain performance improvement on this multimeter. Main outguard processor is also handing keyboard processing, display operation and math 4356a for received data, as well as provide digital interface to computer via GPIB.
Modifications Right after manyal and initial checks, instrument was modified — installed temperature sensor, improved thermal and air insulation around voltage reference zener, and later whole reference module was replaced with DIY triple LMbased module.